Beijing Nubway S&T Development Co., Ltd
Business type: Manufacturer, Exporter
Telephone : 86-10-51206522
Room805,Yabao Road 12#,Chaoyang District, Beijing 100020 China
|Place of Origin:||BEIJING|
|Minimum Order Quantity:||1 SET|
|Packaging Details:||Alumina Box/wooden case|
|Delivery Time:||Within 5 working days|
|Payment Terms:||TT，Western Union,PayPal|
|Supply Ability:||300 SETS PER MONTH|
|Product Name:||Q Switched ND Yag Laser||Pulse Energy:||400mJ-800mJ|
|Pulse Duration:||10ns||Cooling System:||Water Cooling|
Pulsed Q Switched ND Yag Laser 220v With Lcd Screen For Pigmented Lesion
Components of the Unit Description
Principle of Therapy
Specifications and performance Guideline
1. Temperature: 10℃~30℃
2. Relative humidity: ≤75%
Air pressure: 860-1060hpa
Power requirement: AC 220v±10％； 50Hz±2% ;
Water: Distilled water
1. Dimensions: 920× 555 ×1065mm not including joints arm
2. Weight: 100kg
3. Wastage power: 800VA
4. Fuse 220V, 10A
5. Output mode: Q-switch Pulsed output
6. Wavelength: 1064nm, 532nm
7. Pulse Duration: 10±3ns
8. Frequency: 1Hz, 2Hz,5Hz
9. Single Pulse energy: ≥250mJ
10. Safety distance: NOHD 1km. Operator must wear goggles
11. Cooling system: Water cooled
12. Work mode: Continually work with intervals
13. Spot Size: Continually adjustable
Anti indications and side effect
Optical and electrical performance
Laser Skin-care System
1. The system contains dedicate laser generator and electrical components. Heavily Shock and extrusion is not allowed when operating and storing the system. After each treatment, The joints arm should put on the original place. Use the dust-proof cloth to cover the system.
2. Clean the lens of the joints arm with the no water alcohol after each treatment. The lens is at the tip of the joints arm. Revolve the handle to joints arm can get the lens.
Warning: Do not check and clean the lens when the system is on power.
Calibration of the specifications
3 . The calibration of the output:
1) Turn on the system
2) Adjust the light guide system, make the laser pulse can direct at the centre of the reaction interface of the energy calibrator.
3) When the system is at WORK status, Choose lowest Frequency(1Hz) to test the energy of the output.
4 . The main body contains input protective fuse to avoid the system destroyed by the over loaded high voltage and high current of the power supply.
The step of change the broken fuse: The fuse is at the back side of the main body. Loose the fuse cap and take off the broken fuse.
5 . If the environmental temperature change too much, the explosive power will reduce, and at the same time, the thermal efficiency will increase. Then emit the laser to the printing paper. If there is smoke rise from the paper, the operator must ask the technician to check the optical-electrical switch.
6 . When the error value between practical output and expected output is ≥±20%. The calibration
by professional technician is necessary.